The Indialantic buoy was last heard from on 1/10/2016. While we do have a visual on it we have been unable to get out to it to attempt on site troubleshooting. We are currently working on a code upgrade for the transmitter which should help us in determining communication issues in the future and once we can get to the buoy we will swap out transmitters to implement the code upgrade. We appreciate your patience.
We had a great weather window today and were able to return the Indialantic buoy back to service! 1 foot of viz but other than that everything went well.
The Indialantic surf buoy was recovered today, we will repair it as soon as possible and return it to service as quickly as we can. Thank you for your patience.
We lost contact with the Indialantic surf buoy mid-day on 11/24/2015. The issue is related to the same problem that brought down the Cocoa Beach buoy. We will recover the Indialantic buoy at the next available opportunity, repair it and return it to service as quickly as possible. Thank you for your patience.
The Indialantic buoy survived! We went to pay our last respects as soon as we got back beachside from the evacuation and there it was just sitting out there looking pretty. But it did stop transmitting data due to Hurricane Matthew. This buoy will be recovered and repaired as soon as possible. Stay tuned for updates.
Finally had good enough ocean conditions to get the kayak out to assess the damage to the Indialantic buoy and try to retrieve the transmitter. Fortunately the only physical damage was to the antenna mast. The antenna mast had broken off stopping the data transmission and caused the buoy to get flooded. The transmitter is inside a waterproof housing so it is perfectly fine and the battery was fully charged so the solar panels were still functioning as well. The last transmission from the buoy was at 10/6/2016 23:00 EDT at Hs = 12.5 ft and Tp = 9.1 s. The processed data is also stored on the transmitter as a back up and we had hoped to get all the data from the storm off of it but unfortunately the flooding compromised the portion of the data cable inside the buoy. There are a couple of wiring splices inside the buoy that are only splash proof and being submerged in sea water allowed water ingress and has caused intermittent connection. We did get 3 more data points during the storm with the last reading at 10/7/2016 04:30 EDT with Hs = 18.7 ft and Tp = 10.7 s! We also had intermittent data up until today so we know the wavegage is also still working and collecting data. We replaced the broken antenna mast with a dummy antenna mast to keep anything else from getting in and hopefully keep this buoy afloat until we can swap it out with a new buoy. Once we recover it we will also recover the wavegage and get all the data from Hurricane Matthew. Stay tuned as we will publish all the storm data once recovered.
The Indialantic Buoy has been built and is about to enter the full system test period. Once testing is complete in about a week we will begin to look for a suitable weather window to return it to service.
The Indialantic Buoy has been swapped out and data is now streaming to the website. Great conditions and we managed to do this by kayak. Waves were about 1.5ft, water temperature about 72 and almost 10 ft of viz and this is November.
We went out to do maintenance on the Indialantic buoy today but discovered an issue with the transmitter that has forced us to take this buoy down for additional maintenance. The issue should be fixed in the next few days but we will need a good weather window to get back on the water to finish the maintenance.
The communication issue that the Indialantic buoy was experiencing after maintenance has sorted itself out and appears to be working fine with full data return. We will continue to monitor in case the issue reappears.
Weather and waves are still hampering our efforts to restore this buoy to operational status. We will get to it as soon as conditions become favorable to get out to it in a kayak.
We got some good conditions and attempted an in-situ repair that did not work. We replaced the data cable from the wavegauge to the buoy but this did not resolve the issue. We tried resetting the transmitter but to no avail. We are working on the next attempt which will be to replace just the surface buoy. As with replacing the data cable in-situ we have never swapped just the surface buoy before so this is also an experiment. Sorry for the down time and we expect to make the next attempt in late May.
We finally sorted out the issue with the Indialantic buoy. It turns out that during one previous attempt to fix we had a faulty data cable and then on the next attempt we swapped out the cable and wavegauge but unfortunately we did not realize the wavegauge was set up for full-duplex communication while the buoy is set up for half-duplex communication. So they could not talk to each other. But we made the corrections and now have data streaming once again!